LIGO Document G1500623-v6
- This file card and associated related documents are the collection of representative strain sensitivity plots for the aLIGO Interferometers, compared at select times in their commissioning and/or observing process.
This file card shows the latest and greatest strain and displacement sensitivities for the aLIGO interferometers. To see prior sensitivities, to download the ASCII dump of these curves, and to find more details about the detector at that time, follow the tree-structure of related documents at the bottom of this file card.
FOR ASDs OF PREVIOUS RUNS PERFORMANCE, SEE EARLIER VERSIONS OF THIS DOCUMENT.
- This data is merely representative of the detectors performance for ~500 [sec] of data, meant to serve as a snap shot for publication. In reality the detector sensitivity varies as a function of time due to the environment and the health of the instruments subsystems. This variation in sensitivity is on the order of 10% (where percent variation is on the detector sensitivity recast as the distance out to which specific astrophysical source type, under a certain choice cosmological parameters, may be detected).
- The calibration for these sensitivities is only valid between 10 Hz and 5000 Hz, for this the only frequency region in which we have well-quantified the uncertainty.
- The uncertainty in amplitude calibration over this band is roughly 3%. One can find a comparison against the absolute reference for each detectors at select frequencies for this specific time if one drills down to the individual file cards for these traces.
The raw text files in order to reproduce these plots in your own desired format can be found in the other files of the following file cards:
G1902349 -- H1, No subtraction, 36.6 W Input, 2 dB Squeezing
G1902350 -- H1, Calibration line and 60 Hz subtraction, 36.6 W Input, 2 dB Squeezing
G1902351 -- H1, Calibration line and 60 Hz subtraction, non-stationary 60 Hz subtraction, 36.6 W Input, 2 dB Squeezing
G1902348 -- L1, No subtraction, 40 W Input, 2 dB Squeezing
G2000047 -- L1, Calibration line and 60 Hz subtraction, 40 W Input, 2 dB Squeezing
G1902347 -- L1, Calibration line and 60 Hz subtraction, non-stationary 60 Hz subtraction, 40 W Input, 2 dB Squeezing
DCC Version 3.4.1, contact
Document Database Administrators